Sechrist Industries Approved Chamber Cleaners document (P/N 100397 Rev 2): is the most up to date listing of cleaners which have been tested and approved to be utilized on Sechrist monoplace hyperbaric chambers. Specifically, these products have been tested and deemed safe for use on the acrylic cylinder. These are the only cleaning agents that should be used on the Sechrist acrylic cylinders. Since information is becoming available daily, we suggest that your facility keep in contact with the manufacturer of the cleaning agent utilized to determine any new information available about the effectiveness against novel coronavirus. We also suggest that each healthcare facility should utilize the CDC website to review information for Healthcare Professionals and Resources for Healthcare Facilities. Thank you for your cooperation and understanding.
Supply Chain Notification : Sechrist Industries has reviewed if there is any potential impact to our supply chain as well as the production of our respiratory and hyperbaric products. The Sechrist Supply Chain team has been in contact with all of our suppliers and at this time there is no issue with the continuity of items from our suppliers.
Our team is continuing to monitor the situation and we are committed to continuing our supply to you, our valued customer.
Gas Source Selection Pane
The Gas Selection Panel, when used in conjunction with the SECHRIST INDUSTRIES Monoplace Hyperbaric System, Model 2500B, 2800, and H Series will allow the chamber operator to select either oxygen or medical grade air to be supplied to the chamber. The panel incorporates pressure gauges for the oxygen, medical grade air and the selected gas source. The design of the gas selector valve utilizes a “shear type” sealing method, and the Valve vents to ambient. There by preventing interflow between gas supply systems. The panel is also provided with an extra selected gas outlet for ventilator operation.